Place of Origin:
ISO 13485, CE
Total Triiodothyronine(TT3)Rapid Quantitative Test(Fluorescence immunoassay)
1. Limits of detection,No more than 0.50 nmoL/L.
2. Accuracy,The relative deviation from the target value is within ±15%.
3. Precision,The within and between assay coefficient of variations are within 15%.
4. Linear range,Within the linear range (0.50 ~ 10.00nmoL/L), the linear correlation coefficient R≥0.990.
【Storage conditions and validity】
The kit should be stored at 4℃~30℃, out of direct sunlight. It is valid for 18 months. The test card should be used within 15 minutes after unsealing under the environment of 15℃~30℃ and 20% ~ 90% relative humidity.
The production date, batch number and expiration date are shown in the outer package of the product.
|Test cards||25||It is composed of fluorescent pad (coated with fluorescent labeled T3 monoclonal mouse antibody), nitrocellulose membrane (coated with T3 monoclonal mouse antibody and Goat anti mouse IgG antibody), absorbent paper and backing|
|Sample diluent||25(200μL/tube)||Phosphate buffer|
|ID card||1||With specific stand curve file|
The components in different batches of kits cannot be used interchangeably.
NIR-1000 dry fluoroimmunoassay analyser produced by WWHS Biotech. Inc.
WWHS NIR-1000 dry fluoroimmunoassay analyser
Package and Delivery
Packing Details : Standard carton package or as demand.
Delivery Details : 5-15days after payment
We support a variety of transportation modes, including land
transportation, sea transportation, air transportation, DHL, Fedex,TNT and other international logistics express.
Send your inquiry directly to us