Place of Origin:
Diagnostic Kit for Interleukin-6(Immunochromatographic assay)
The kit is used for quantitative determination of Interleukin-6 (IL-6) in human whole blood, plasma and serum.
Interleukin 6 is a member of the family of interleukins. It plays an important role in the inflammatory response and is a sensitive indicator for the early diagnosis of acute infection.
The kit is a one-step chromatographic sandwich immunoassay designed for the quantitative measurement of IL-6. The IL-6 antigen in the sample was first bound with the conjugated compound of fluorescent labeled IL-6 monoclonal antibody, then moved and combined with another IL-6 monoclonal antibody fixed on the nitrocellulose membrane, and the double antibody sandwich complex was formed at the detection line of the cellulose nitrate membrane. The quantitative detection results were obtained by NIR-1000 dry fluoroimmunoassay analyser.
|Test cards||25||It is composed of fluorescent pad (coated with fluorescently-labeled IL-6 monoclonal antibody), nitrocellulose membrane (coated with IL-6 monoclonal antibody and Goat anti mouse IgG antibody), absorbent paper and backing.|
|Sample diluent||25 (0.3mL/ tube)||Phosphate buffer|
|ID card||1||With specific stand curve file|
The components in different batches of kits cannot be used interchangeably.
【Storage conditions and validity】
The kit should be stored at 4℃~30℃, out of direct sunlight. It is valid for 18 months. The test card should be used within 15 minutes after unsealing under the environment of 15℃~30℃ and 20% ~ 90% relative humidity.
The production date, batch number and expiration date are shown in the outer package of the product.
NIR-1000 dry fluoroimmunoassay analyser produced by WWHS Biotech. Inc.
The normal reference value is less than 10pg/mL in this assay. It is strongly recommended that each laboratory should determine its own normal and abnormal values.
【Interpretation of test results】
【Limitation of rest method】
4. When the concentration of IL-6 in the sample is less than 10000ng/mL, there is no hook effect.
1. Limits of detection
No higher than 5.00 pg/mL.
The relative deviation from the target value is limited to ±15.0%.
The within and between assay coefficient of variations are within 15.0%.
4. Linearity range
Within the linear range (5.00~ 4000.00)ng/mL, the linear correlation coefficient R≥0.990.
1. The kit can be used for in vitro diagnosis only.
2. Test card and buffer solution are single-use and they cannot be reused.
3. Please check the integrity and validity of the kit package before use, and then open the package. When it is stored at low temperature, it should be restored to room temperature (15℃ ~ 30℃) before opening the package for use. The reagents with damaged inner package and beyond the validity period cannot be used.
4. Take the test card out of the aluminum foil bag and carry out experiment in 15min. Do not place it in the air for a long time to avoid dampness.
5. It is required to strictly comply with the requirements for sample collection and storage. If the sample is turbid, please centrifuge and precipitate it before use.
6. The kit used should be disposed of as latent infective material, and all samples, reagents and latent contaminants should be disinfected and disposed of according to relevant local regulations.
|cat#.||Product item||Specimen||Reaction Time||Measure Range||Clinical Range||Itended Use|
|10||CRP / hs-CRP||WB/Serum/Plasma||3min.||0.5-200mg/L||CRP<10mg/L,hs-CRP<1mg/L||nonspecficity inflammatory marker.|
|13||CRP+SAA||WB/Serum/Plasma||5min.||same with single item||same with single item||inflammation&infection.|
1. Please thoroughly read the specification before test. Frozen test card and sample should be placed at room temperature (15-30) ℃ for at least 30min before use.
2. Start NIR-1000 dry fluoroimmunoassay analyser and verify quality control according to the specification.
3. The quality of reagent is controlled using internal quality control product (not provided) of the company and result should be controllable.
4. Take out the test card from the aluminum foil bag and use it within 15min
5. Place the test card on a clean horizontal table top and label it.
6. Take 10μL of sample and add it into HbA1c sample buffer (1.00mL). Then, mix the solution evenly, take 100μL of the solution and add it into the well.
7. Insert the test card into NIR-1000 dry fluoroimmunoassay analyser, keep time for 10min and press 【Instant test】. The analyser will judge and read the test result automatically. Or press 【Fixed time test】 to keep time for 10min automatically. The analyser will judge and read the test result automatically and display it in the screen.
WWHS NIR-1000 dry fluoroimmunoassay analyser
Q1: Are you trading company or manufacturer ?
A1: We are manufacturer.
Q2: Do you provide samples ?
A2: Yes,we do provide samples.If you need it, please contact us.
Q3: How about the quality?
A3: We have the best professional engineer and strict QA and QC system to ensure you get high quality products from us.
Send your inquiry directly to us