|Place of Origin:||CN|
|Certification:||ISO 13485, CE|
|Minimum Order Quantity:||1000|
|Price:||US 1.50-2.99 Kit|
|Packaging Details:||Colorful Paper Box|
|Delivery Time:||7 Days|
|Payment Terms:||L/C, T/T|
|Supply Ability:||20000 Kits per Week|
|Product Name:||D-Dimer Rapid Test Kit||Storage:||Room Temperature|
|Sample Buffer:||25 Vials||Manufacturer:||WWHS BIOTECH INC|
|Usage:||Medical Hospital Clinical||Format:||Cassette|
|Supply Ability:||10 Million A Year||Package:||25 Tests|
|Keywords:||High Sensitive, Super Stability|
D-Dimer Rapid Quantitative Test(Fluorescence immunoassay)
This kit is used for quantitative determination of D-dimer in human whole blood and plasma.
The principle of immunofluorescence chromatography was applied to the kit. The D-dimer antigen in the sample was first bound with the conjugated compound of fluorescent labeled D-dimer monoclonal antibody, then moved and combined with another D-dimer monoclonal antibody fixed on the nitrocellulose membrane, and the double antibody sandwich complex was formed at the detection line of the cellulose nitrate membrane. The quantitative detection results were obtained by NIR-1000 dry fluoroimmunoassay analyser.
|Test card||25||It is composed of fluorescent pad (coated with fluorescent labeled D-dimer monoclonal mouse antibody), nitrocellulose membrane (coated with D-dimer monoclonal mouse antibody and Goat anti mouse IgG antibody), absorbent paper and backing|
|Sample buffer||25(300μL/tube)||Phosphate buffer|
|ID card||1||Record the standard curve information of this kit|
The components in different batches of kits cannot be used interchangeably.
【Storage conditions and validity】
The test card should be stored at 4℃~30℃, dry, dark and no freezing. It should be stored in sealed aluminum foil bag and valid for 12 months. The test card should be returned to room temperature (15℃~30℃) before opening. It should be used within 15 minutes after unsealing under the environment of 15℃~30℃ and 20% ~ 90% relative humidity.
The production date, batch number and expiration date are shown in the outer package of the product.
NIR-1000 dry fluoroimmunoassay analyser produced by WWHS Biotech. Inc.
By measuring 268 healthy people aged between 19 and 79, statistical analysis shows that the reference interval is less than 500 ng/ml. According to the characteristics of the local population, the reference intervals of the laboratories were established.
【Interpretation of results】
【Limitations of methods】
1. Analysis sensitivity
No more than 40ng/ml.
The relative deviation from the target value is within ±15%.
The within and between assay coefficient of variations are within 15%.
4. Linear range
Within the linear range (40 ~ 10000ng/ml), the linear correlation coefficient R≥0.990.
1. This kit is only used for in vitro diagnosis.
2. The test card and sample diluent are disposable and cannot be reused.
3. Please check the integrity and validity of the kit package before use, and then open the package. When it is stored at low temperature, it should be restored to room temperature (15℃~30℃) before opening the package for use. The reagents with damaged inner package and beyond the validity period cannot be used.
4. The requirements of specimen collection and storage should be strictly observed. If the specimen is turbid, it should be centrifuged and discarded before use.
5. The used kits should be treated as potential infectious substances, and all samples, reagents and potential pollutants should be disinfected and treated according to the relevant local regulations.
|WWHS Assay list|
|cat#.||Product item||Specimen||Reaction Time||Measure Range||Clinical Range||Itended Use|
|1||cTnI||WB/Serum/Plasma||12min.||0.1-40ng/ml||<0.3ng/ml||several heart diseases including myocardial infarction and heart failure.|
|2||Myo||WB/Serum/Plasma||12min.||5-400ng/ml||<58ng/ml||acute myocardial infarction (AMI) in early stage.|
|3||CK-MB||WB/Serum/Plasma||12min.||1-200ng/ml||<5ng/ml||acute myocardial infarction (AMI) in early stage.|
|heart failure .|
|5||D-Dimer||WB/Plasma||10min.||40-10000ng/ml||<500ng/ml||disseminated intravascular coagulation (DIC),deep vein thrombosis (DVT),pulmonary embolism (PE), myocardial infarction, cerebral infarction, etc.|
|6||cTnI+Myo+CKMB||WB/Serum/Plasma||12min.||same with single item||same with single item||Triple marker of myocardial infarction.|
|7||ST2||WB/Serum/Plasma||10min.||10-400ng/ml||<35ng/ml||heart failure .|
|8||Lp-PLA2||WB/Serum/Plasma||10min.||10-900ng/ml||<175ng/ml||Risk evaluation of ACS and atherosclerotic ischemic stroke patients.|
|9||S100-β||WB/Serum/Plasma||10min.||0.05-10ng/ml||<0.2ng/ml||Cerebral infarction, cerebral injury.|
Interpretation of Test Results
1. The kit can be used for auxiliary test only. If test result is abnormal, retest timely and judge combined with clinical symptoms.
2. For samples whose cTnI concentration is lower than 0.1ng/mL and higher than 40ng/mL, test result is “＜0.1ng/mL” and “＞40ng/mL” respectively.
A procedural control is included in the test. A colored line appearing in the control region (C) is the internal procedural control. It confirms sufficient specimen volume and correct procedural technique.
Control standards are not supplied with this kit; however, it is recommended that a positive control and a negative control control be tested as a good laboratory practice to confirm the test procedure and to verify proper test performance.
Contact Person: sales
Address: Rm 505, 1st Building, Shenzhen Biomedicine Innovation Industrial Park, No. 14th, Jinhui Road, East Jinxiu Road, Kengzi Street, Pingshan District, Shenzhen
Factory Address:Rm 505, 1st Building, Shenzhen Biomedicine Innovation Industrial Park, No. 14th, Jinhui Road, East Jinxiu Road, Kengzi Street, Pingshan District, Shenzhen