Place of Origin:
ISO 13485, CE
Diagnostic Kit for C-reactive Protein (Immunochromatographic assay)
It is used for quantitative detection of C-reactive protein (CRP) in human serum, plasma and whole blood. CRP is mainly used as a non-specific inflammatory index.
The kit is used the principle of fluorescence immunochromatography. The CRP antigen in the sample first combines with the fluorescent labeled CRP monoclonal antibody conjugate, and then continues to move and combine with another CRP monoclonal antibody fixed on the nitrocellulose membrane to form a double antibody sandwich immune complex at the detection line of nitrocellulose membrane. The quantitative detection results are obtained by NIR-1000 dry fluoroimmunoassay analyser.
|Test card||25||It is composed of fluorescent pad (coated with fluorescent labeled CRP monoclonal antibody), nitrocellulose membrane (coated with CRP monoclonal antibody and Goat anti mouse IgG antibody), absorbent paper and backing|
|Sample diluent||25||Phosphate buffer|
|ID card||1||Record the standard curve information of this batch of reagents|
The components in different batches of kits cannot be used interchangeably.
【Storage conditions and validity period】
4℃ - 30℃, dry, dark, no freezing, sealed in aluminum foil bag, valid for 18 months. The test card should be returned to room temperature (15-30)℃ before use, and should be used within 15 minutes after unsealing under the environment of temperature (15-30) ℃ and relative humidity (20% - 90%).
NIR-1000 dry fluoroimmunoassay analyser produced by WWHS Biotech. Inc.
2. Venous blood was collected according to routine laboratory methods, and hemolysis was avoided as much as possible in the process of treatment.
3. After clinical samples were collected, the detection was completed within 4 hours at room temperature (15-30) ℃.The whole blood sample can be stored for 24 hours at (2-8)℃ without freezing; Serum and plasma samples can be stored for 7 days at 2-8℃;Serum and plasma samples - 20℃ for 30 days.
4. The sample must return to room temperature (15-30)℃ before testing. The frozen samples should be completely thawed, rewarming and mixed evenly before use, and repeated freezing and thawing should be avoided.
5. Do not test samples with severe hemolysis, severe lipidemia and jaundice.
1. Before the test, please read the instructions completely. If the test card and sample are stored in cold storage, they should be balanced at room temperature (15-30)℃ for not less than 30min before use.
2. Start NIR-1000 dry fluoroimmunoassay analyser according to the instruction manual of the instrument, and carry out quality control verification according to the instruction manual of the instrument(Note: the reagent has been calibrated in advance, and the calibration curve parameters of each batch of reagent have been stored in the ID card. The test can be carried out only after the quality control is passed. Otherwise, the cause should be found out before testing.)
3. Remove the test card from the aluminum foil bag and use it within 15 minutes.
4. Place the test card on a clean horizontal table and mark it horizontally.
5. Add 5μL sample to sample diluent (1500μL), mix well and add 100μL diluted sample to the sampling hole.
6. Insert the test card into dry fluoroimmunoassay analyser, press the “timing test” key, automatically time for 3 minutes, automatically judge the test results, and display the quantitative results on the screen. Or insert the test card into the analyzer after 3 minutes, and press the “Instant test” key, and the instrument will automatically interpret the results.
252 healthy people aged 19-80 were tested. The upper limit of CRP reference value was 10mg/L at 95th percentile. Reference interval: CRP<10mg/L.
It is suggested that each laboratory should establish its reference range according to the characteristics of local population.
【Interpretation of test results】
1. This reagent is only used for auxiliary detection. If the test results are abnormal, it should be reviewed in time and judged in combination with clinical symptoms.
2. For samples with CRP concentration lower than 0.5mg/L and higher than 200mg/L, the test results were reported as "<0.5mg/L " and ">200mg/L" respectively.
【Limitations of test methods】
1. This kit is only used to detect human serum / plasma / whole blood samples.
2. Due to the limitations of serological methods for antigen and antibody reactions, the results obtained by detection cannot be used as the only basis for clinical diagnosis, but should be evaluated together with all existing clinical and experimental data.
3. The contents of triglyceride, hemoglobin and bilirubin in the samples were not more than 20 mg/ml, 10 mg/ml and 0.4 mg/ml, respectively, and the relative deviation of the test results was not more than 0.5%±10%.
4. When the concentration of C-reactive protein was less than 400 mg/L, there was no hook effect.
5. When the concentration of human anti mouse in the sample is less than 50ng/ml, HAMA effect will not be produced.
6. When the RF concentration in the sample is less than 2000 IU/ml, the relative deviation of the detection results is not more than±10%.
1. Detection limit: No higer than 0.5mg/L.
2. Accuracy: the relative deviation from the target value is not more than±10%.
3. Repeatability: coefficient of variation (CV) should be no more than 10%.
4. Inter batch difference: the relative range (R) between batches should be no more than 15%.
5. Linear range: within the specified linear range of 0.5mg/l-200mg/l, the linear correlation coefficient R ≥ 0.990.The absolute deviation of the linear range in [0.5-5] mg/L is not more than ±5 mg/L, the linear range is (5-200] mg/L, the relative deviation is less than±10%.
1. The kit is only used for in vitro diagnosis.
2. The test card is disposable and cannot be reused.
3. Please check the integrity and validity of the kit package before use, and then open the package. When it is stored at low temperature, it should be restored to room temperature (15-30) ℃ before opening the package for use. The reagents with damaged inner package and beyond the validity period cannot be used.
4. The test card should be taken out from the aluminum foil bag and tested within 15 minutes to avoid being placed in the air for a long time and causing damp.
5. The requirements of specimen collection and storage should be strictly observed. If the specimen is turbid, it should be centrifuged and discarded before use.
6. The kit contains products from animals. The qualified information of animal source and health status cannot guarantee the existence of infectious pathogens. Therefore, it is suggested that these products should be treated as potential infectious substances, and all samples, reagents and potential pollutants should be disinfected and treated according to the relevant local regulations.
7. Hematocrit too high or too low may affect the results of whole blood test, it is recommended to use other detection methods for verification.
|cat#.||Product item||Specimen||Reaction Time||Measure Range||Clinical Range||Itended Use|
|10||CRP / hs-CRP||WB/Serum/Plasma||3min.||0.5-200mg/L||CRP<10mg/L,hs-CRP<1mg/L||nonspecficity inflammatory marker.|
|13||CRP+SAA||WB/Serum/Plasma||5min.||same with single item||same with single item||inflammation&infection.|
1. Plasma, serum and whole blood can be used as samples. The whole blood should be collected in a tube containing heparin, citrate or EDTA as the anticoagulant. If the serum procedure is used, collect blood in a tube without anticoagulant and allow clotting. Hemolyzed samples should not be used.
2. Venous blood was collected according to routine laboratory methods to avoid hemolysis.
3. It is highly recommended to use fresh samples instead of keeping the samples at room temperature for a long time. After samples were collected, the detection should be completed within 4 hours at room temperature (15℃~30℃). The whole blood sample can be stored at 2℃~8℃ for 24 hours. Plasma and serum samples can be stored at 2℃~ 8℃ for 7 days, -20℃for 30 days.
4. Before testing, the sample should return to room temperature (15℃~30℃). The frozen samples should be completely thawed, rewarming and mixed evenly before use. Repeated freeze-thaw cycles should be avoided.
WWHS NIR-1000 dry fluoroimmunoassay analyser
A procedural control is included in the test. A colored line appearing in the control region (C) is the internal procedural
control. It confirms sufficient specimen volume and correct procedural technique.
Control standards are not supplied with this kit; however, it is recommended that a positive control and a negative control control be tested as a good laboratory practice to confirm the test procedure and to verify proper test performance.
1. What is the MOQ?
Generally if you choose the different products, our minimum order qty. also will be different. One sample available.
2. What about the delivery time?
The goods need 7-21 work days to be made upon deposit.
3. Does your product have any warranty?
Yes, we offer 6 months limited warranty for our products.
Damage due to misuse, ill treatment and unauthorized modifications and repairs are not covered by our warranty.
4. What's your payment method?
We usually accept the payment methods include PayPal, TT, and Western Union.
50% deposit in advance and balance before shipping. Buyer can choose which payment ways that you accept.
PayPal is most useful and safe, please don’t refuse PayPal fee when you choose PayPal.
5. What's your shipping method?
We provide comprehensive shipping methods.
For small quantity orders we ship by DHL Air-Express, or EMS/TNT/UPS/FEDEX Express service, it is safe and fast.
For large quantity orders we ship by Air transportation or sea transportation, we can also ship order to buyer’s cargo agent in China.
Send your inquiry directly to us