Place of Origin:
Diagnostic Kit for S100β protein(Immunochromatographic assay)
The kit is used for quantitative determination of S100β in human whole blood, serum or plasma. Clinically, by detecting the expression of S100β protein, we can judge the degree of brain injury and evaluate the prognosis of patients.
The Diagnostic Kit for S100β is a one-step chromatographic sandwich immunoassay designed for the quantitative measurement of S100β. The S100β antigen in the sample was first bound with the conjugated compound of fluorescent labeled S100β monoclonal antibody, then moved and combined with another S100β monoclonal antibody fixed on the nitrocellulose membrane, and the double antibody sandwich complex was formed at the detection line of the cellulose nitrate membrane. The quantitative detection results were obtained by NIR-1000 dry fluoroimmunoassay analyser.
|Test cards||25||The product consists of fluorescent pat (coated with fluorescently-labeled S100β monoclonal antibody), nitrocellulose membrane (coated with S100β monoclonal antibody ), absorbent paper and PVC soleplate.|
|Sample diluent||25 (200µL/ tube)||Tris-HCl buffer|
|ID card||1||With specific stand curve file|
The components in different batches of kits cannot be used interchangeably.
【Storage Conditions and Validity】
The kit should be stored at 4℃~30℃, out of direct sunlight. It is valid for 18 months. The test card should be used within 15 minutes after unsealing under the environment of 15℃~30℃ and 20% ~ 90% relative humidity. The sample buffer is disposable and used immediately after opening the cap.
The production date, batch number and expiration date are shown in the outer package of the product.
NIR-1000 dry fluoroimmunoassay analyser produced by WWHS Biotech. Inc
1,Plasma, serum and whole blood can be used as samples. The whole blood should be collected in a tube containing citrate as the anticoagulant. If the serum procedure is used, collect blood in a tube without anticoagulant and allow clotting. Hemolyzed samples should not be used.
2,Venous blood was collected according to routine laboratory methods to avoid hemolysis.
3,It is highly recommended to use fresh samples instead of keeping the samples at room temperature for a long time. After samples were collected, the detection should be completed within 4 hours at room temperature (15℃~30℃). The whole blood sample can be stored at 2℃~8℃ for 48 hours. Plasma and serum samples can be stored at 2℃~ 8℃ for 7 days, -20℃for 30 days.
4,Before testing, the sample should return to room temperature (15℃~30℃). The frozen samples should be completely thawed, rewarming and mixed evenly before use. Repeated freeze-thaw cycles should be avoided.
1,Before the test, please read the instructions completely. If the test card and sample are stored in cold storage, they should be balanced at room temperature (15-30)℃ for not less than 30min before use.
2,Start NIR-1000 dry fluoroimmunoassay analyser and correctly select the corresponding sample type on the instrument.
3,Take out the ID card, make sure that the batch number of the ID card is consistent with that of the test card, and insert the ID card into the ID card port of the instrument.
4,Take out the test card from the aluminum foil bag and use it within 15 minutes.
5,Place the test card on a clean horizontal table and mark it horizontally.
6,Mix 100 µL of urine sample with 200µL of sample diluent. Apply 100 µL of diluted samples to the well of the test card.
7,Insert the test card into NIR-1000 dry fluoroimmunoassay analyser and click the “instant test” button to read the results at 10 minutes after addition of samples.
S100β normal reference value is less than 0.20ng/mL. It is strongly recommended that each laboratory should determine its own normal and abnormal values.
【Interpretation of test results】
1,The kit can be used for auxiliary test only. If test result is abnormal, retest timely and judge combined with clinical symptoms.
2,For samples whose S100β concentration is lower than 0.05ng/mL and higher than 10.00ng/mL, test result is “＜0.05ng/mL” and “＞10.00ng/mL” respectively.
【Limitation of test method】
1,This kit is only used to detect human serum/plasma/whole blood samples
2,Due to the limitations of immunoassay methods of antigen and antibody reaction, the results cannot be used as the only basis for clinical diagnosis, but should be evaluated with all the existing clinical and experimental data.
3,The triglyceride content in the sample does not exceed 15mg/mL, the hemoglobin content does not exceed 5mg/mL, the bilirubin content does not exceed 0.5mg/mL, the content of neuron-specific enolase does not exceed 200ng/mL, the content of glial fibrillary acidic protein does not exceed 5ng/mL, the content of IL-6 does not exceed 200pg/mL, the content of tumor necrosis factor does not exceed 200pg/mL, the content of albumin does not exceed 60mg/mL, the content of fibroblast growth factor does not exceed 20ng/mL, the cholesterol does not exceed 10mg/mL, the content of total protein does not exceed 120mg/mL, and the relative deviation of the measurement results does not exceed ±15.0%.
4, When S100β concentration of samples reaches 40.00ng/mL, there is no hook effect.
5,HAMA effect was not produced when the concentration of human anti rat in the sample was less than 50ng/ml.
6, When RF concentration of samples is less than 2000IU/mL, relative deviation of test result is limited to ±15.0%.
7,For samples exceeding the linearity range, test cannot be conducted after dilution.
1. Limits of detection ,No higher than 0.05ng/mL
2. Accuracy,The relative deviation to the target value is limited to ±15.0%.
3.Precision,The within and between assay coefficient of variations are within 15%.
4.Linear range,Within the linear range [0.05,10.00]ng/mL, the linear correlation coefficient R≥0.990
1. The kit can be used for in vitro diagnosis only.
2. Test card and buffer solution are single-use and they cannot be reused.
3. Please check the integrity and validity of the kit package before use, and then open the package. When it is stored at low temperature, it should be restored to room temperature (15℃ ~ 30℃) before opening the package for use. The reagents with damaged inner package and beyond the validity period cannot be used.
4. Take the test card out of the aluminum foil bag and carry out experiment in 15min. Do not place it in the air for a long time to avoid dampness.
5. It is required to strictly comply with the requirements for sample collection and storage. If the sample is turbid, please centrifuge and precipitate it before use.
6. The kit used should be disposed of as latent infective material, and all samples, reagents and latent contaminants should be disinfected and disposed of according to relevant local regulations.
|cat#.||Product Item||Specimen||Reaction Time||Measure Range||Clinical Range||Intended Use|
|1||cTnI||WB/Serum/Plasma||12min.||0.1-40ng/ml||<0.3ng/ml||several heart diseases including myocardial infarction and heart failure.|
|2||Myo||WB/Serum/Plasma||12min.||5-400ng/ml||<58ng/ml||acute myocardial infarction (AMI) in early stage.|
|3||CK-MB||WB/Serum/Plasma||12min.||1-200ng/ml||<5ng/ml||acute myocardial infarction (AMI) in early stage.|
|heart failure .|
|5||D-Dimer||WB/Plasma||10min.||40-10000ng/ml||<500ng/ml||disseminated intravascular coagulation (DIC),deep vein thrombosis (DVT),pulmonary embolism (PE), myocardial infarction, cerebral infarction, etc.|
|6||cTnI+Myo+CKMB||WB/Serum/Plasma||12min.||same with single item||same with single item||Triple marker of myocardial infarction.|
|7||ST2||WB/Serum/Plasma||10min.||10-400ng/ml||<35ng/ml||heart failure .|
|8||Lp-PLA2||WB/Serum/Plasma||10min.||10-900ng/ml||<175ng/ml||Risk evaluation of ACS and atherosclerotic ischemic stroke patients.|
|9||S100-β||WB/Serum/Plasma||10min.||0.05-10ng/ml||<0.2ng/ml||Cerebral infarction, cerebral injury.|
1. Advanced Quality: We have strict quality control system and enjoy good reputation in
2. Sincerly Service: We treat clients as friend and aims at building long term business
3. Affordable Price: We always ensure high price performance to benefit our customers.
4. Timely Delivery: Your order shall be shipped once finished
1. What is the MOQ?
Generally if you choose the different products, our minimum order qty. also will be different. One sample available.
2. What about the delivery time?
The goods need 7-21 work days to be made upon deposit.
3. Does your product have any warranty?
Yes, we offer 6 months limited warranty for our products.
Damage due to misuse, ill treatment and unauthorized modifications and repairs are not covered by our warranty.
4. What's your shipping method?
We provide comprehensive shipping methods.
For small quantity orders we ship by DHL Air-Express, or EMS/TNT/UPS/FEDEX Express service, it is safe and fast.
For large quantity orders we ship by Air transportation or sea transportation, we can also ship order to buyer’s cargo agent in China.
Send your inquiry directly to us